Portable Dry Fluorescence Immunoassay Analyzer MKF-201
● Software name: Portable dry fluorescence immunoassay analyzer
● Detection time: V 1.0
● Service period: 3 years
Used in conjunction with an appropriate reagent for the quantitative analysis of the substance to be measured in human samples.
Automatic immunofluorescence detection system is based on the principle of photoelectricity detection, which needs to be used with special fluorescent immune reagent. It is widely used in disease detection, medical care, epidemic monitoring, food inspection, quarantine and other fields. Apply to various kinds of medium and small medical institutions and rescue centers.
Working principle of the instrument
During the detection, when the substances to be measured in the sample form immune complexes with fluorescent labeled antibodies and undergo a chromatographic process, they are solidified in the detection area and control area of the dry fluorescence immunoassay reagent card, respectively. The excitation light source is irradiated to the detection area and quality control area of the reacted reagent card, the cured fluorescent complexes are excited to emit fluorescence, and The fluorescence is converted to current by the photodiode. The current intensity is proportional to the fluorescence intensity. The portable dry fluorescence immunoassay analyzer automatically calculates the concentration of the sample to be measured by analyzing the current intensity.
Scope of application
The portable dry fluorescence immunoassay analyzer needs to be used with a series of specific reagents based on fluorescence immunochromatography. It is used only by medical professionals for in vitro diagnostic experiments. It can be used in medical institutions.
System components and main structures
After opening the package, first check whether the components are missing or damaged according to the following standard configuration list.
Note: If any components are missing or damaged, please contact Jiangsu Medomics Medical Technology Co. Ltd. and local sales representative. Please refer to Chapter 7 of the instructions for more contact information.